RS PRO IDM63N Handheld LCD Digital Multimeter, AC Voltage, Continuity, DC Current, DC Voltage, Diode Test, Resistance
- RS Stock No.:
- RS PRO
- COO (Country of Origin):
RS PRO IDM63N Pocket Digital Multimeter
RS PRO IDM63N high-performance digital multimeter measuring AC-DC voltage, electrical current, resistance, continuity and diode test. This meter has an LCD display where the measurement readouts can be viewed.
RS PRO range of handheld and bench digital multimeters are RS approved meaning that they are great value for money and high quality you can rely on.
Features and Benefits
3200 Count digital display
65 segment analogue bar graph
Full Autoranging with manual override
Continuity and Diode Test
Auto Power Off
Fast continuity beeper (0.2 msec)
Protective Holster with tilt stand
800-hour battery life
Display hold function
Conforms to IEC1010-1 600 V category III
What does it measure?
DC and AC Voltages - up to 600 V
DC Current - up to 3000μA
Resistance and continuity checking - up to 30 MΩ
Diode - open circuit voltage up to 3.3 V
Ergonomically designed slim shape with one handed operating knob and buttons. It just fits into the palm of your hand.
The multimeter is supplied with a protective holster that includes a built-in stand and convenient test lead storage. This holster provides an extra degree of protection and convenience. Wrap the leads around the holster to store the test probes.
Batteries can be easily replaced by removing the battery door for battery replacement. Designed for users convenience without opening the bottom case.
What does it include?
Pair of Test Leads
A digital multimeter is a handy test tool that measures electrical values such as voltage, current, and resistance. A DMM is used by technicians in the following areas:
Electrical and electronic industries
When using any electrical test equipment it is advised that you refer to the users manual for proper working operating procedures and safety precautions in order to avoid potential safety hazards.
What is the difference between an analogue and a digital multimeter?
The main difference is the display. A digital multimeter will show the results as numbers on a screen (usually a LED or LCD screen) whereas the analogue version has a needle to show the value.
|Absolute Maximum DC Current Measurement||3mA dc|
|Absolute Maximum AC Voltage Measurement||600V ac|
|Absolute Maximum Resistance Measurement||30MΩ|
|Absolute Maximum DC Voltage Measurement||600V dc|
|Functions Measured||AC Voltage, Continuity, DC Current, DC Voltage, Diode Test, Resistance|
|Model Number p||IDM63N|
|DC Current Measurement Resolution||0.1µA dc|
|AC Voltage Measurement Resolution||1mV ac|
|Resistance Measurement Resolution||100mΩ|
|DC Voltage Measurement Resolution||0.1mV dc|
|Maximum Operating Temperature||+50°C|
|Best AC Voltage Accuracy||±1.5% + 5 Digits|
|Battery Life||800 h|
|Best Resistance Measurement Accuracy||±1% + 4 Digits|
|Safety Category Level||CAT III|
|Best DC Current Accuracy||±1% + 2 Digits|
|Best DC Voltage Accuracy||±0.5% + 2 Digits|
|Safety Category Voltage||600V|
|Battery Type||AAA 1.5 V, AM4, IEC LR03|
|Dimensions||80 x 36 x 165mm|
|Minimum Operating Temperature||0°C|